- Open Access
A novel method of blind bedside placement of postpyloric tubes
Critical Care volume 22, Article number: 62 (2018)
We read with great interest the recent report on blind bedside postpyloric placement by Lv et al. . Their methods were proven to be safe and effective in intensive care units. Although our placing procedure is similar to that reported by the authors, the choice of a postpyloric tube and the patient’s position requires further improvement.
In our center, we use a 130-cm long transpyloric tube with a guide wire (CH10–130, inner diameter 2.0–2.1 mm, Flocare, Nutricia Ltd, Wuxi, China; Fig. 1a) rather than the spiral feeding tube used by Lv et al. The Flocare tube that we used has several advantages compared with the spiral tube. First, the Flocare tube is inexpensive, approximately $22, in China, whereas the spiral tube costs approximately $71. Expense is extremely important in developing countries and at one-third less compared with the latter, our technique is easier to implement in hospitals and areas with limited resources. Second, the Flocare tube that we used has two side holes near its tip (Fig. 1b); therefore, it is less likely to be blocked compared with the spiral tube. Third, the guide wire is shorter in length compared with the Flocare tube; thus, the rigid tip would not damage the digestive tract during the placing procedure.
The patient’s position before placing the postpyloric tube also requires further improvement. In our procedure, the patient is placed in a right decubitus position at 30–45° after gastric placement is accomplished, followed by a postpyloric placement. According to our experience, in this position the tip of the Flocare tube falls to the pylorus ostium by gravity, which may increase the placement success rate.
The results of our data (unpublished) analysis confirmed our improved methods (Fig. 2). From December 2016 to December 2017, 44 patients underwent postpyloric tube placement using our novel techniques. In total, 38 cases (86.4%) were successful, and 33 cases (75.0%) were successful at the first attempt. The success rate and first-time success rate of our placement techniques were better than those described in the study by Lv et al. The median time of our procedure was 13 (8.5–16) minutes, and the median insertion length was 100 (93.5–110) cm. These values are similar to the results of Lv et al. and other previous reports [1,2,3].
Considering the less expensive tube and better first-time success rate, our novel blind bedside postpyloric placement may be easier to implement worldwide, and we look forward to collaborating with the authors and other colleagues.
Lv B, Hu L, Chen L, Hu B, Zhang Y, Ye H, Sun C, Zhang X, Lan H, Chen C. Blind bedside postpyloric placement of spiral tube as rescue therapy in critically ill patients: a prospective, tricentric, observational study. Crit Care. 2017;21(1):248.
Rollins CM. Blind bedside placement of postpyloric feeding tubes by registered dietitians: success rates, outcomes, and cost effectiveness. Nutr Clin Pract. 2013;28(4):506–9.
Kohata H, Okuda N, Nakataki E, Itagaki T, Onodera M, Imanaka H, Nishimura M. A novel method of post-pyloric feeding tube placement at bedside. J Crit Care. 2013;28(6):1039–41.
This work was supported in part by grants from the National Natural Science Foundation of China (number 81701881), and the Nanjing Medical Science and Technology Development Foundation (number YKK15098, number YKK17102).
Availability of data and materials
Ethics approval and consent to participate
Consent for publication
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
See related research by Lv et al., https://ccforum.biomedcentral.com/articles/10.1186/s13054-017-1839-2
About this article
Cite this article
Sun, J., Wang, X. & Yuan, S. A novel method of blind bedside placement of postpyloric tubes. Crit Care 22, 62 (2018). https://doi.org/10.1186/s13054-018-1986-0