- Meeting abstract
- Open Access
Nitrogen dioxide (NO2) production for different doses of inhaled nitric oxide (NO) during mechanical ventilation with different tidal volumes using two prototypes for the administration of NO
© Current Science Ltd 1997
- Published: 1 March 1997
- Nitric Oxide
- Mechanical Ventilation
- Tidal Volume
- Nitrogen Dioxide
- Control Mechanical Ventilation
To test the amount of NO2 production during mechanical ventilation with different concentrations of inhaled NO (0.5-90 ppm) and different tidal volumes (VT 100–1000 ml) using two prototypes for NO inhalation during mechanical ventilation.
The Servo 300 NO-A prototype (Siemens-Servotek, Solna, Sweden) and the NO-domo prototype combined with an Evita 1 ventilator (Dräger AG, Lübeck, Germany) were tested for controlled mechanical ventilation at an FiO2 of 1.0 with different tidal volumes (VT 100, 150, 200, 250, 300, 400, 500,... 1000 ml) using an artificial test lung with a filling volume of 3 1. Different NO concentrations of 0.5, 1,10, 20, 40 and 90 ppm (NOin) were applied. In the Servo prototype, NO is mixed into the inspiratory gas stream inside the ventilator and electrochemical electrodes for NO/NO2 are mounted in a mixing chamber behind the expiratory valve of the respirator. The NO-domo prototype admixes NO into the inspiratory gas stream immediately behind the respirator's inspiratory outlet and the electrochemical measurement is mounted 30 cm distal from that point in the inspiratory limb. NO2 concentrations were measured by chemiluminescence in the inspiratory limb of the respiratory tubing immediately before the y-piece (NO2 insie for the Servo, NO2 innodo for the NO-domo). Furthermore, for the Servo 300 NO-A an additional chemiluminescence measurement was mounted in the expiratory mixing chamber at the same point where the electrochemical electrodes measure (NO2 exsie).
The NO2 values for the different measurements are shown in the figure for the different NO concentrations. Each data point for a given NO concentration reflects a stepwise increasing VT from 100-1000 ml.