- Poster presentation
- Open Access
Glucometer accuracy and implications for clinical studies
Critical Care volume 16, Article number: P173 (2012)
Elucidating links between glycemic control and clinical outcome requires reliable discrimination between groups with different target blood glucose (BG) cut-offs. Point-of-care glucometers are commonly used, but lower accuracy means BG errors will impact classification and thus outcome analyses. This study reanalyses a BG control trial with an error model of a typical glucometer to assess the impact of sensor errors on interpretation of trial results.
BG profiles from 301 patients (stay >24 hours) from the SPRINT trial with BG measurements (n = 25,000) using the Arkray SuperGlucocard II GT-1630. A model of sensor bias and variance (CV 2.7 to 3.5%, regression: y = 3.92 + 0.97x) was used to estimate possible 'true' BG profiles from measured BG and repeated 100 times for each patient. The defined cut-off for 'good' control for a patient was ≥70% of BG in 72 to 126 mg/dl (cTIB ≥0.7), and 'poor' as <70% (cTIB <0.7), based on original observed clinical BG. The number of true BG profiles that resulted in misclassification between 'good' and 'poor' control for a patient was recorded over all Monte-Carlo runs. The maximum change in true and observed BG mean and standard deviation were used to evaluate potential worst-case scenarios.
Good control was clinically measured in 76% of patients (24% with cTIB <0.7). Of these, 83% of 'good' and 64% of 'poor' control would never be misclassified over all 100 runs due to sensor error. A total of 91% (good) and 87.5% (poor) could be misclassified 10% of the time. Patients with cTIB near 0.7 were more likely to be misclassified when accounting for glucometer error. Hence, a deadband around the cut-off would reduce this misclassification. If 'good' cut-off was cTIB ≥0.5 (95% of clinical patients) then 100% correct classification was 97% for good control patients, but fell to 40% of poor control patients. The median largest difference in observed and true mean BG across patients was -54 mg/dl (90th percentile: -21 mg/dl) and the standard deviation was 3.2 mg/dl (90th percentile: 1.8 mg/dl).
Glucometers can distinguish between patients that received good and poor BG control but risk of misclassification rises for patients nearer cut-offs. Reliable classification to associate with outcomes relies on the control protocol and cut-off choice to achieve sufficient separation between groups so that device errors do not result in significant misclassification confounding the results. A deadband around cut-off values to eliminate patients at high risk of misclassification may be required.
About this article
Cite this article
Le Compte, A., Pretty, C., Shaw, G. et al. Glucometer accuracy and implications for clinical studies. Crit Care 16 (Suppl 1), P173 (2012). https://doi.org/10.1186/cc10780
- Blood Glucose
- Glycemic Control
- Blood Glucose Control
- Blood Glucose Measurement
- Sensor Error