Pulse pressure analysis: to make a long story short

Pulse pressure analysis algorithms are commonly used to measure cardiac output and to allow for the rational titration of therapy in critically ill patients. The ability of these algorithms to accurately track changes in stroke volume (and cardiac output) is thus very important. Most of the currently available algorithms can provide robust data so long as there is no fundamental change in the vasomotor tone (arterial compliance or impedance). If the tone changes significantly, for instance with vasodilatation or vasoconstriction, then the data become less robust. For this reason, unless there is a mechanism for compensating for changes in vasomotor tone, these algorithms are best used only over short time periods in order to get the most accurate and precise data on changes in cardiac output.

Many authors have discussed the importance of measuring cardiac output and then titrating therapy according to these measurements in patients in the operating theatre [1,2] and intensive care environments [3]. Indeed, in some circumstances these measurements have led to changes in therapy that, in themselves, have been associated with improvements in outcomes [3]. Th e 'art' or 'science' of measuring this variable is therefore rightly given signifi cant airplay in the ongoing literature of our specialty [4].
Th ere are nowadays many devices available that pur port to measure cardiac output. Th ese include methodologies based on indicator dilution or thermodilution, Doppler principles, the Fick technique and also pulse pressure analysis. Th e pulse pressure analysis techniques have become increasingly popular due to the rising number of companies now marketing these devices [4]. It is incumbent on us as practicing clinicians to understand the similarities and diff erences between these devices so that we can ensure that we use techniques that we can rely upon to be accurate and precise in the clinical environ ment and also then integrate with therapies that are benefi cial to our patients.
If we step back and look carefully at how these tools are used, then we would purport that there are two diff erent scenarios that could be discussed. Th e fi rst scenario is where a snapshot of the circulatory status is required. Th is needs an accurate and precise measurement in order to provide useful information [5][6][7]. Th e second scenario is where clinical interventions are titrated against changes in cardiac output -for instance, with a passive leg raise [8,9] or volume challenge [2]. In this scenario it is less relevant that we have an accurate and precise measurement, although it is more important that we can track the changes in the underlying signal reliably [10]. On the whole, the pulse pressure analysis techniques for estimating cardiac output are better placed at helping us with this second scenario than the fi rst. In order to have an accurate and precise measurement, the relationship between arterial pressure and central impedance needs to be clarifi ed and this usually means having to make an independent measurement as impedance is notoriously diffi cult to measure. Most companies therefore market these devices combined with another method of measuring cardiac output to calibrate the pulse pressure algorithm at baseline for this problem -commonly with either transpulmonary thermodilution or lithium (indicator) dilution techniques.
On a beat to beat basis pulse pressure provides a very good surrogate of changes in stroke volume. As the time interval lengthens, however, this relationship becomes less robust as the vascular tone will change, thereby adversely infl uencing this signal. Th e same holds true for the measurement of changes in stroke volume and/or cardiac output from pulse pressure tracking techniques. Over time many of the competing infl uences on the systemic vasculature will alter -level of preload, compliance, arterial resistance, and so on. Th is makes the assumption that changes in the arterial pressure signal directly relate to changes in fl ow less robust. On a beat to beat basis many of the marketed technologies will provide reliable information. Unfortunately, these tools are rarely used over a beat to beat basis and are more commonly used

Abstract
Pulse pressure analysis algorithms are commonly used to measure cardiac output and to allow for the rational titration of therapy in critically ill patients. The ability of these algorithms to accurately track changes in stroke volume (and cardiac output) is thus very important. Most of the currently available algorithms can provide robust data so long as there is no fundamental change in the vasomotor tone (arterial compliance or impedance). If the tone changes signifi cantly, for instance with vasodilatation or vasoconstriction, then the data become less robust. For this reason, unless there is a mechanism for compensating for changes in vasomotor tone, these algorithms are best used only over short time periods in order to get the most accurate and precise data on changes in cardiac output. over a period of time that may be 30 minutes or perhaps over an hour. If we look at the variety of methodologies used for giving a fl uid challenge we can see this all too vividly. Many authors give the fl uid over a 30 to 60 minute time window [11]. After 60 minutes it is quite possible that the vascular tone has changed signifi cantly, thereby raising the question as to whether the change in fl ow estimated from the pressure signal is real or artefactual.
In order to understand this problem a number of authors have investigated these techniques under changing circulatory conditions. In an elegant study, Marquez and colleagues [12] demonstrated that the LiDCOplus algorithm, when compared against aortic fl ow probes, was able to track changes in stroke volume in response to a venous occlusion, although there tended to be an under estimation at higher values. Yamashita and colleagues [13,14] assessed how the precision of the algorithms was maintained under therapeutic vasodilatation with prostaglandin E1 during cardiac surgery. Th ey tested the LiDCO TM plus and the pulse contour method of the PiCCOplus versus the intermittent thermodilution of the pulmonary artery catheter. Th ese studies suggested that after signifi cant haemodynamic change (vasodilatation), the algorithms may underestimate the cardiac output and therefore not give a reliable estimate in the change of the signal. More recently, Monnet and colleagues [1] assessed how the PiCCOplus and the Vigileo (v1.10) handle vasoconstriction induced by infusion of nor epinephrine. Th ey concluded that the Vigileo algorithm was less able to track the changes in cardiac index during these situations. A further important consideration from all of these studies is that each algorithm, or algorithm update, will behave diff erently and will require inde pendent validation. Th is can be seen in the meta-analysis published by Mayer and colleagues [15] looking at the new and older versions of the Vigileo algorithms where dramatically diff ering levels of accuracy and precision were seen.
It seems clear that if these devices are to be used to be able to track changes in cardiac output induced by changes in preload, then much care must be taken to ensure that in addition there are no major infl uences from altered vascular tone. Th e only way of ensuring this is to make the time interval between measurements shortperhaps minutes rather than hours. If we want to assess the circulation over longer time intervals, then a measure ment independent of pulse pressure analysis needs to be included to compensate for these changes in vascular tone. When designing methodologies for assessing the response to a passive leg raise [8], an end expiratory occlusion [16], a Valsava manoeuvre [17] or a fl uid challenge [2] this message needs to be understood.
Perform the intervention quickly and the monitor should be able to track the change reliably and the correct interpretation should be made.